The inputs can be from thermocouples or platinum resistance thermometers. CP1W-TS/TS and CPM1A-TS/TS Temperature Sensor Units. Omron CP1W-TS 51 available from 14 distributors. Explore Industrial Control on Octopart: the fastest source for datasheets, pricing, specs and availability. A. CP1W-TS CompoBus/S I/O Link Unit. 8 inputs and 8 outputs. CP1W -SRT A. P20E-EN+CP1L+ Seite 2 Donnerstag,
|Published (Last):||13 November 2012|
|PDF File Size:||12.52 Mb|
|ePub File Size:||5.68 Mb|
|Price:||Free* [*Free Regsitration Required]|
Started by arrowsmith7431 Jan Posted 31 Jan Everything is brand new, so I’m quite sure I am the problem here If someone have an answer and maybe even an programming example to upload I would really appreciate it.
CP1W-TS Omron Expansion Temp – More Control
PT sensor test 3. Your dip-switches and rotary switch seem OK. The ladder is not a problem, since you don’t need any ladder at all to read values in your 4 channels. CIO2, 3, 4 and 5.
I can’t help with this, I only worked with thermocouples, but everything else seem OK so maybe you should try to connect otherwise. Posted 31 Jan edited. Posted 2 Feb The bits are allocated from A Unit nearest the CPU Unit.
Omron CP1 PLC Temperature Sensor Module 4-Point Platinum Resistance Thermometer Input
Temperature Sensor Units are allocated two bits each. Please tell us the following: Value of Auxiliary Word A Provide picture of dip switch and rotary switch.
Also rotary position for 0 is not directly up. Check rotary switch to be slightly to the right from up. Cp1 everything and powered up, and suddenly it worked perfectly without doing any changes Explained the situation and he checked by Teamviewer and we came to the conclusion that it was hard to do anything else than let it be. Stuffed everything back in the box and took it back to the workshop. Powered up and guess what, problem is back and there are no values.
Didn’t know if i should be mad or happy, but i decided to be happy and started ruling out the potential culprits and.
You need to be a member in order to leave a comment. Sign up for a new account in our community. Share this post Link to post Share on other sites. You ccp1w not mention, what values are you seeing in your CIO2, 3, 4 and 5?
Cp1w-ts Wholesale, Cp1w-ts Suppliers – Alibaba
CIO2, 3, 4 and 5 What about the wiring? I will do my wiring explanation by both words and some pictures I have all 4 sensors hooked up to the TS I measured the sensor wires to make a desicion which wire is which.
Switched places for all 3 wires of first sensor, no luck. Edited 31 Jan by arrowsmith Your wiring is perfect. With an open-circuit detection, it is not reflected in word A Value of Auxiliary Word A Provide picture of dip switch and rotary switch.
Powered up and guess what, problem is back and there are no values to read again. It was the xp1w that caused me this problem! Thanks to everyone involved here for taking your time to help me! Create an account or sign in to comment You need to be a ts02 in order to leave a comment Create an account Sign up for a new account in our community. Register a new account.
Sign in Already have an account? Go To Topic Listing Omron.
Sign In Sign Up.